Dionex SRD-10抑製再生劑檢測器 074395保護您的抑製器和使用外部水或化學再生模式下，盡一切運行計數。
保護您的抑製器和使用外部水或化學再生模式下，盡一切運行計數。 Thermo Scientific戴安SRD-10抑製再生劑檢測器，如果抑製再生流動停止或減緩，防止對抑製或浪費的離子色譜法（IC）運行損壞自動禁用洗脫液泵。
Dionex SRD-10抑製再生劑檢測器 074395
|SRD-10 Suppressor Regenerant Detector||074395|
The SRD-10 Suppressor Regenerant Detector is a stand-alone device that monitors liquid flow to a suppressor. If the flow is restricted or stops, the SRD-10 automatically disables the eluent pump via a TTL command, thus preventing irreversible damage being done to the suppressor.
The SRD-10 is intended for use with these applications:
An SRS® 300 Self-Regenerating Suppressor that is operated in the external water mode
An MMS™ 300 MicroMembrane™ Suppressor that is operated in the chemical regenerant mode
In the external water mode of operation, an external water supply is pumped into the SRS regenerant channels. A constant current is applied across the electrodes in the SRS to electrolytically regenerate the device. It is possible for the IC (ion chromatography) system to continue to operate and for current to be applied to the SRS even after external water flow has stopped or the water supply is depleted. Unless the current is turned off, this will quickly cause irreversible damage to the SRS.
In the chemical regenerant mode of operation, running out of regenerant will not damage the MMS, but it will adversely affect the chromatographic run.
In either of these scenarioses, the SRD-10 will shut off the eluent pump, stopping the run. In the case of the SRS, the eluent pump shut off will switch off the suppressor.